This is an old revision of the document!
Mentor: S. Steinke
A ceilometer measures the cloud base height, i.e. the ceiling in aviation. It is based on the light detecting and ranging (LIDAR) principle, i.e. it sends laser pulses and measures the backscattered light. From travel time the distance of the back scattering object can be determined. From strength and attenuation of the backscattered signal the backscatter coefficient is determined. The instrument identifies cloud bases by the strong increase of the backscatter coefficient. If the lower clouds are transparent it can detect up to three distinct cloud layers. We use the backscatter profile for an estimate of boundary layer height.
- If you are interested in receiving data please contact the instrument mentors: S. Steinke
|Pulse Power||16 W typical|
|Pulse energy||1.6μJ ±20%1)|
|Average Power||8.9 mW|
|Pulse Width||100 ns|
|Repetition Rate:||5.57 kHz|
|Number of pulses:||65,5362)|
|Range resolution||50ft = 30m (see here)|
|Range||0…25,000 ft. (0…7.5 km)|
|Laser Divergence||± 0.53×0.75 mrad (edge×diagonal3))|
|Field-of-View Divergence||± 0.66 mrad|
|Size||760 x 280 x 245 mm|
|Power Consumption||365 W|